Save up to 80% on pre-owned Alaris/Cardinal Imed Gemini PC-4 infusion pumps.
Request a Quote for this product now.
12.0″ x 14.75″ x 7.5″ (30.5 cm x 37.5 cm x 19.1 cm)
26.5 lbs. (12.0 kg)
90 – 132 VAC, Current draw 0.1A nominal (0.36A maximum), 10 watts, fused at 0.4A, 50/60 Hz, 3 wire, single phase
With new, fully charged batteries, approximately 4 hours with All four channels infusing as primary infusion at 125 ml/hr before a “BATTERY DISCHARGED” message.
Less than 100 microamps
Note: Recommended maximum rate in the Controller Mode is 500 ml/hr.
1 ml/hr if set rate is 1 ml/hr or above; or set rate, if rate is 0.9 ml/hr or below
Container height ±12″ (±30 cm)
10±2 psi (517 ± 103 mmHg) for rates above 30 ml/hr. For rates of 30 ml/hr and below, the occlusion pressure will be less than 10 psi, an is rate-dependent.
25 – 517 mmHg in 25 mmHg increments to 500 mmHg.
INFUSION COMPLETE – KVO, EMPTY CONTAINER – KVO, MULTIDOSE COMPLETE, DELAYED START COMPLETE , LOW BATTERY, Maintenance Mode, Replace Battery, PAUSE, Paused, Powering Down in X Seconds, Panel/keypad locked, Panel/keypad Unlocked, LOW FLOW, START = XX:XX. SECONDARY, DOSE N = XX:XX and ANESTHESIA MODE – PAUSE